Standards Related to G39 – G G Standard ASTM Standards Related to Corrosion Testing of Metals · Annual Book of ASTM Standards Online: Hello everyone, I am looking for ASTM G39 – 99() Standard Practice for Preparation and Use of Bent-Beam Stress- Corrosion Test Specimens. Please. ASTM G39 – 99() – 弯曲梁应力腐蚀试验试样制备和使用标准规程.
|Published (Last):||23 July 2016|
|PDF File Size:||5.50 Mb|
|ePub File Size:||3.86 Mb|
|Price:||Free* [*Free Regsitration Required]|
In sheet-gage, bent-beam specimens the de? NOTE 8—If the test is to be xstm in an electrolyte, the spacer shall be made of the same material as the specimen or of an electrically nonconducting material such as glass, ceramic, and so forth to prevent galvanic action between specimen and spacer.
Stress calculations by this practice are not applicable to plastically stressed specimens. Stress calculations by this practice are not applicable to plastically stressed specimens.
The two inner supports shall be located symmetrically around the midpoint between the outer supports.
It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
Laboratory test data for carbon and low-alloy steels have been found to correlate with? A convenient stressing jig is xstm in Fig. The other quantities are given in Inspection of Specimens The equations must not be used above the yield strength of the material. Once cracking has initiated, the state of stress at the tip of the crack as well as in uncracked areas has changed, and therefore the known or calculated stress or strain values discussed in this method apply only to the state of stress existing before initiation of cracks.
3 & 4 POINT BEND FIXTURES – Cortest
Advantages of such a notched specimen include the probable localized cracking in the notch and an acceleration of failure. The stresses in awtm engineering structure can be varied between operational design stresses and residual stresses from heat treatment or fabrication.
However, unless directly related to practical conditions of usage, misleading failures may ensue.
This can be accomplished by mounting the specimen holders on appropriate racks and exposing the entire rack to the environment. For plate ashm the bent-beam specimen is atsm difficult to use, because more rugged specimen holders must be built to accommodate the specimens. Support the specimen at the ends and bend the specimen by forcing a screw equipped with a ball astn knife-edge tip against it at the point halfway between the end supports in a fashion shown in Fig.
For instance, the loading bolt can be supplanted by a spring or deadweight arrangement to change the mode of loading. This equation can be solved by computer, by trial and error, or by using a series expansion of the sine function. For plate material the bent-beam specimen is more difficult to use because more rugged specimen holders must be built to accommodate the specimens.
Corrosion 14, 11 p. The inch-pound units in parentheses are provided for information. Note 1-It is the nature of these methods that only the applied stress can be calculated.
A specimen thickness of about 0. Current edition approved March 1, This relationship can be used as a simple check to ensure that the maximum stress does not exceed the proportional limit. The bent-beam specimens are designed for testing at stress levels below ast, elastic limit of the alloy. It should also be noted that presence of voluminous corrosion products may obscure cracks, thus making a careful examination mandatory. It is the responsibility of the user of this standard to establish appropriate safety and health asgm and determine the applicability of regulatory limitations prior to use.
The common parameter in these equations is the modulus k of the elliptic integrals. Your comments are invited either for revision of this asgm or for additional standards and should be addressed to ASTM International Headquarters.
T39 2—The double-beam specimen, more fully described in Since stress-corrosion cracking is a function of the total stress, for critical applications and proper interpretation of results, the residual stress before applying external stress or the total elastic stress after applying external stress should be determined by appropriate nondestructive methods, such as X-ray diffraction 1.
Referenced Documents purchase separately The documents listed below are referenced within the subject standard but are not provided as part of the standard. The actual bending stress there will be greater by a concentration factor dependent on the notch geometry, than the minimal test stress, and generally, may be expected to be in the range of plastic stain.
This excludes corrosion-reduced sections aztm fail by fast fracture. A number in 3g9 indicates the year of last reapproval. By testing a set of specimens at a series of stress levels, the stress dependence of alloys can be assessed. The applied stress is determined from the size of the specimen and the bending de? By proper choice of holder and insulator materials, however, many plastics can be used, especially in short-time tests.
3 & 4 POINT BEND FIXTURES
The specimen is therefore unsuitable for studying the effects of different axtm stresses on stress-corrosion cracking or for studying variables which have only a minor effect on cracking. The following paragraphs give relationships used to calculate the maximum longitudinal stress in the outer?
Care must be exercised not to contaminate cleaned specimens. We expertize in translation of technical documents and also deliver procurement service of Chinese versions. Stress calculations by this practice are not applicable to plastically stressed specimens. The thickness of the specimen is usually dictated by the mechanical properties of the material and the product form available.