to reading the User’s Manual. Please read the Accutrend. Plus User’s Manual carefully before using the meter for the first time. The Accutrend Plus meter is used. The Accutrend Plus system, one meter delivering both total cholesterol and glucose test results, can with Accutrend Plus may help employers improve employee productivity, improve .. 15 Cholestech LDX System User Manual. Inverness. View and Download Cobas Accutrend Plus user manual online. Accutrend Plus Blood Glucose Meter pdf manual download.
|Published (Last):||16 May 2010|
|PDF File Size:||12.95 Mb|
|ePub File Size:||8.21 Mb|
|Price:||Free* [*Free Regsitration Required]|
Display Of Results Quality control recommendation Display of results When measurement is finished, the result is displayed. Press manuao Set button to save the displayed year.
Quality control recommendation After switching on the instrument, check the following displays: Lift the lid upwards to remove it from the instrument. The display then automatically switches to the mode for setting the current month. Solution Power llus instrument off and on again.
If you leave the cap off the container of test strips. Safety Information Safety information Protection against infection There is a potential risk of infection. Installation, use and maintenance of the Accutrend Plus instrument is accutrenr full responsibility of the user. Performing a measurement Prepare the lancing device by inserting a new lancet. Regularly check that all display elements are functioning correctly to prevent misinterpretations due to a defective display.
Page 79 Performing a measurement Assigning additional information accytrend a measurement: For each test parameter, separate control solutions are available. Do not touch the application area with the finger. Corrections can only be made by repeating the settings.
Refer to the insert of the respective test strip. The imprinted barcode contains lot-specific information which is read and stored in the instru- C Test strip bottom side. This information is stored and must therefore only be read once per test strip container.
Performing a measurement The flashing drop symbol above the strip symbol now instructs you to apply the blood. Performing a measurement A flashing arrow now instructs you to open the measurement chamber flap to apply the blood.
Flap open displayed during measurement The measurement chamber flap is not closed properly. Page 64 Performing a measurement Therefore, clean the accuutrend regularly and immediately after it becomes dirty.
When the instrument detects that the test strip is inserted.
Use only the following items for cleaning: Performing a measurement Sample material Fresh capillary blood is used as sample material. The display then automatically switches to the mode for setting the time format. Instrument settings The default month flashes.
Cobas Accutrend Plus User Manual
This column contains the illustration of a display This column contains information relating to the screen. Memory After the usual display test has been performed when switching on the instrument, the last stored measured value is displayed. Quality control recommendation Starting the measurement 11 Close the measurement chamber flap.
In the professional, multi-patient sector you should use a suitable single-use lancing device to prevent cross contamination. When the test strip reaches the correct position, you will hear two beeps short — Cleaning If needed, you can rinse the strip guide separately from the instrument under warm running water.
COBAS ACCUTREND PLUS USER MANUAL Pdf Download.
If the error still persists, the instrument is defective. Setting The Date Instrument settings Setting the date With the next three settings, you first enter the year, then the month and finally the day. Consult instructions for use. If you are measuring glucose, perform the plausibility check described on the next page. The instrument must be powered off before you can activate the set mode.
Temperature The instrument or ambient temperature does not fall within the acceptable range for the selected test. Page 10 Contents Product specifications Operating conditions and technical data Note that all handling of blood samples represents a risk of infection. All instructions which require you to perform an action, in addition to very important information, appear on a colored background. Error Messages Error messages Error messages In certain circumstances error messages may appear on your display.
Page 78 Performing a measurement Flagging the measurement as a control: Error messages Display Error meaning and description Error: Page 3 The packaging material, the identification plate of the instrument and the manual may contain the following symbols or abbreviations which are listed below with their meaning: Moving backwards is not possible.